The CR10 had 12 single-ended (six differential) analog input channels, three switched excitation outputs, eight input/output control ports, two pulse counting channels, and a serial I/O port. It provided sensor measurement, timekeeping, data reduction, data/program storage and control functions. Up to 29,908 data points were stored in it's internal memory.
CR10s manufactured after 1 December 1993 (SN>19273) contained an internal jumper that supported use of 2 or 4 kbytes program memory (appropriate PROM also required).
|-NOTE-||Note: Additional specifications are listed in the CR10 Specifications Sheet.|
|Analog Input Channels||12 single-ended (six differential)|
|Switched Excitation Outputs||3|
|Input/Output Control Ports||8|
|Pulse Count Channels||2|
|Serial I/O Port||1|
|Memory||Stores up to 29,908 data points in internal memory.|
A typical field-based CR10 system consisted of:
CR10 peripherals such as the CR10KD and CR10TCR, as well as sensors and data retrieval peripherals, were added to complete the system.
Number of FAQs related to CR10: 2
A practical maximum is to connect one multiplexer per every two control terminals on the data logger. Control terminals can be shared between multiplexers to increase the number of connected multiplexers. Sharing terminals, however, requires more complex wiring and programming. Users who would like to connect more than one multiplexer per every two control terminals are advised to contact a sales or support engineer at Campbell Scientific for assistance.
The CR10 and CR10X can use the same wiring panel. The original CR10 wiring panel design was improved by doing the following: