The CR1000X is our flagship data logger that provides measurement and control for a wide variety of applications. Its reliability and ruggedness make it an excellent choice for remote environmental applications, including weather stations, mesonet systems, wind profiling, air quality monitoring, hydrological systems, water quality monitoring, and hydrometeorological stations.
The CR1000X is a low-powered device that measures sensors, drives direct communication and telecommunications, analyzes data, controls external devices, and stores data and programs in onboard, nonvolatile storage. The electronics are RF-shielded by a unique sealed, stainless-steel canister. A battery-backed clock assures accurate timekeeping. The onboard, BASIC-like programming language, common to all contemporary Campbell Scientific data loggers, supports data processing and analysis routines.Read More
The CR1000X is a low-powered device designed to measure sensors, drive direct communication and telecommunications, analyze data, control external devices, and store data and programs in on-board, non-volatile storage. The electronics are RF-shielded and glitch-protected by a unique sealed, stainless-steel canister. A battery-backed clock assures accurate timekeeping. The on-board, BASIC-like programming language—common to all Campbell Scientific data loggers—supports data processing and analysis routines.
The CR1000X wiring panel includes two switchable 12 V terminals, analog grounds dispersed among 16 analog terminals, and unpluggable terminal blocks for quick deployment.
|-NOTE-||Additional specifications are listed in the CR1000X Specifications Sheet.|
|Operating Temperature Range||
|Case Material||Powder-coated aluminum|
|Analog Inputs||16 single-ended or 8 differential (individually configured)|
|Pulse Counters||10 (P1 to P2 and C1 to C8)|
|Voltage Excitation Terminals||4 (VX1 to VX4)|
|Maximum Source/Sink Current||
|Data Storage Ports||microSD|
|Switched 12 Volt||2 terminals|
|Digital I/O||8 terminals (C1 to C8) configurable for digital input and output. Includes status high/low, pulse width modulation, external interrupt, edge timing, switch closure pulse counting, high-frequency pulse counting, UART, RS-232, RS-485, SDM, SDI-12, I2C, and SPI function. Terminals are configurable in pairs for 5 V or 3.3 V logic for some functions.|
|Input Limits||±5 V|
|Analog Voltage Accuracy||
|Power Requirements||10 to 18 Vdc input|
|Real-Time Clock Accuracy||±3 min. per year (Optional GPS correction to ±10 µs)|
|Internet Protocols||Ethernet, PPP, CS I/O IP, RNDIS, ICMP/Ping, Auto-IP(APIPA), IPv4, IPv6, UDP, TCP, TLS, DNS, DHCP, SLAAC, SNMPv3, NTP, Telnet, HTTP(S), FTP(S), SMTP/TLS, POP3/TLS|
|Communication Protocols||CPI, PakBus, SDM, SDI-12, Modbus, TCP, DNP3, UDP, NTCIP, NMEA 0183, I2C, SPI, and others|
|Warranty||3 years (against defects in materials and workmanship)|
|Battery-backed SRAM for CPU Usage & Final Storage||4 MB|
|Data Storage||4 MB SRAM + 72 MB flash (Storage expansion of up to 16 GB with removable microSD flash memory card.)|
|Idle Current Drain, Average||< 1 mA (@ 12 Vdc)|
|Active Current Drain, Average||
23.8 x 10.1 x 6.2 cm (9.4 x 4.0 x 2.4 in.)
Additional clearance required for cables and leads.
|Weight||0.86 kg (1.9 lb)|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
|LNLINUX||Version 4.5 or higher|
|LoggerLink||Version 1.6 or higher|
|LOGGERNET||Version 4.5 or higher|
|PC200W||Version 4.5 or higher|
|PC400||Version 4.5 or higher|
|RTDAQ||Version 1.3 or higher|
|Short Cut||Version 4.0 or higher|
With several terminal types, the CR1000X is compatible with nearly every available sensor, including analog (both voltage and current), thermocouples, serial, SDI-12, pulse, and frequency sensors.
The CR1000X is compatible with all our CDMs, multiplexers, vibrating-wire interfaces, terminal input modules, and relays.
The CR1000X communicates with a PC via direct USB port, 10/100baseT Ethernet port, multidrop modems, short-haul modems, phone modems (land line, digital cellular, and voice-synthesized), RF telemetry, and satellite transmitters (HDR GOES, Argos, Meteosat, Iridium, and Inmarsat).
Data can be viewed on the CR1000KD Keyboard Display, the CD100 Mountable Display with Keyboard, an iOS or Android device (requires our free LoggerLink app), CD295 DataView II Display, or other third-party devices.
The CR1000X and its power supply can be housed in any of our standard enclosures.
Any external 12 Vdc source can power the CR1000X datalogger. Campbell Scientific power supplies commonly used with the CR1000X are the BPALK, PS150, and PS200.
Campbell Scientific proprietary software is required to program, configure, and communicate with the CR1000X datalogger. Various software options are available to meet a breadth of application needs. At a minimum, Short Cut and PC200W are needed. Consult with a Campbell Scientific Support and Implementation Engineer to determine the best software option.
Number of FAQs related to CR1000X: 6
Expand AllCollapse All
The CR1000X is not compatible with PC208W.
The CR1000 and CR1000X will be sold concurrently for no less than two years. We anticipate the CR1000 will be retired in December of 2019 and will be supported until December of 2029.
When idle, the CR1000, CR6, and CR1000X consume less than 1 mA @ 12Vdc. Similar to the CR6, the CR1000X has a much faster processor that requires more power when up and running. As such, there will be higher current draws during active measurements, serial communications, or when plugged into a PC via USB or Ethernet.
It may be helpful to think of the CR6 and CR1000X as being built on the same "platform."
Return the CR1000X to Campbell Scientific for repair. For assistance obtaining a Returned Material Authorization (RMA), follow the instructions on the Repair and Calibration page.
Although there are significant additions to the CR1000X program instructions that are not available in the CR1000, in most cases, you can load your program written for the CR1000 to a CR1000X with minor instruction changes. The most notable of the minor instruction changes is in the analog measurement instructions. For more information, see the "Replacing Your CR1000 Data Logger with a CR1000X: What You Should Know" blog article.