El sensor CS300 mide la totalidad de luz solar The CS300 measures total sun and sky solar radiation for solar, agricultural, meteorological, and hydrological applications. Its spectral range of 300 to 1000 nanometers encompasses most of the shortwave radiation that reaches the Earth's surface. This pyranometer connects directly to our dataloggers. Its output can be measured by all of our dataloggers.Read More
The CS300 uses a silicon photovoltaic detector mounted in a cosine-corrected head to provide solar radiation measurements for solar, agricultural, meteorological, and hydrological applications. Calibrated against a Kipp & Zonen CM21 thermopile pyranometer, the CS300 accurately measures sun plus sky radiation for the spectral range of 360 to 1120 nm. Sensors calibrated to this spectral range should not be used under vegetation or artificial lights.
The standard output is 0.2 mV per W m-2, which provides a signal of 200 mV in full sunlight (1000 W m-2). All of our dataloggers, including the CR200(X) series, can measure this output.
|Spectral Range||360 to 1120 nm (wavelengths where response is 10% of maximum)|
|Measurement Range||0 to 2000 W/m2 (full sunlight ≈1000 W/m2)|
|Absolute Accuracy||±5% for daily total radiation|
|Calibration Factor||5 W/m2/mV|
|Cosine Correction Error||±5% at 75° zenith angle; ±2% at 45° zenith angle|
|Temperature Response||0.04 ± 0.04% per °C|
|Response Time||< 1 ms|
|Long-Term Stability||< 2% per year|
|Operating Temperature Range||-40° to +70°C|
|Relative Humidity Range||0 to 100%|
|Output Sensitivity||0.2 mV/W/m2|
|Diameter||2.4 cm (0.9 in.)|
|Height||2.5 cm (1.0 in.)|
|Weight||65 g (2.3 oz) with 2-m (6.6-ft) lead wire|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
Accurate measurements require the sensor to be leveled using a 18356 leveling fixture. This leveling fixture incorporates a bubble level and three adjusting screws. The 18356 mounts to a crossarm or a tripod or tower mast using the CM225 mounting stand. The CS300 should be mounted away from all obstructions and reflective surfaces that might adversely effect the measurement.
One differential analog input channel per probe is required.